Questions and answers on CSM products.
CANape automatically uses the installed CSMconfig with the highest version number.
The CANape driver expects exactly defined settings with regard to the network configuration. If an XCP gateway is to be exchanged in an existing CANape project, the new gateway must use identical IP address settings like the replaced gateway. The different serial numbers of the two devices will cause no problem, since the communication between CANape and the modules is identified exclusively via the IP address.
To set the correct connection parameters (IP address), open the "XCP Gateway Configuration" dialog in CSMconfig and adapt it.
Note: If the IP address of the old gateway is not known, there are two ways to find it out.
The CSM INCA AddOns are licensed software. If you do not yet have a license, please contact your personal account manager or email@example.com.
If you already have a license and are only looking for the installation package, please contact your personal account manager or CSM Service and Support.
A complete list of CAN AD modules with TEDS functionality can be found here: Measurement modules with TEDS support
The STGMM 6 CAN and ECAT modules support TEDS (CSM-proprietary template).
The integration of the CAN interfaces into CSM INCA AddOn CAN is based on the BOA driver library of INCA.
This supports all CAN interfaces which are also supported by the BOA installation on your PC. Additionally, ETAS CAN interfaces can be used.
For a complete list of all supported CAN interfaces of your BOA driver library, please contact ETAS.
Currently not all CAN interfaces available in CSMconfig are supported by CSMview. CAN interfaces from the following manufacturers are currently available for CSMview:
Note: BOA driver-based interfaces are not supported.
All module housing dimensions are listed on the CSM website.
1000V voltage inputs and the associated signal cables can be identified by the color coding.
1. On the cable:
2. On the module:
To use ETAS interfaces by CSMconfig, the ETAS BOA driver library must be installed. The required drivers are supplied with the installation of INCA, but can also be downloaded directly from the ETAS download portal.
In order for the ETAS interfaces to be recognized by CSMconfig, the BOA interface library must be selected in the program settings.
Note: Not all ETAS interfaces use the BOA interface. Such interfaces cannot be addressed by CSMconfig despite selection of the BOA interface library.
The AddOns behave like AddOns from different manufacturers. Therefore it is possible to install and use them at the same time.
The AddOn CAN is started via the respective ETAS interface, the AddOn ETH via the Ethernet interface.
Information on cables and general CAN or ECAT accessories can be found here.
The CSMcan can be used with the following software products:
The CSMcan is not supported in CANape and cannot therefore be used as an interface.
The Vector Ethernet interfaces (e.g. VX0312, VN5640) are treated in INCA like an USB-to-Ethernet adapter. The interfaces are therefore not found via the hardware search, but must be created manually as an Ethernet system.
The CSM INCA AddOn is then automatically called via the external hardware search and the connected gateway and the measuring modules are searched. The measurement modules are then configured in the INCA AddOn.
The CSM measurement modules are integrated in the INCA hardware environment. Select the Ethernet system via the menu item "Insert" and integrate the XCP gateway.
(Settings for the IP addresses are to be executed via the target server)
The connected measuring modules can be found via the external hardware search or directly in the AddOn. The measurement modules are configured exclusively in the AddOn. Before starting the measurement, the entire measurement setup must be initialized in INCA.