Frequently asked questions on configuration software CSMconfig.
A list of the interfaces which are supported by CSMconfig can be found in the help under the keyword "Supported CAN interfaces". To use the intefaces you have to install manufacturer's drivers first.
Depending on which driver library you have activated in the CSMconfig program settings, different interfaces can be used:
The following interfaces are currently supported by CSMconfig:
If BOA is selected as the active interface library, CSMconfig supports all CAN interfaces that are also supported by the BOA installation on your PC. For a complete list of all supported CAN interfaces of your BOA driver library, please contact ETAS.
The offset correction via adjustment in CSMconfig takes place in the DBC/A2L file and not in the module. This means that new scaling values are stored in the DBC/A2L file after the offset adjustment has been carried out. If an offset correction is desired during the measurement, it is better to use a function of the measurement software.
The only exception: Remote zero adjustment via CAN message with particular revsions of CAN ADMM8 pro/ADMM8 pro2 .
In the channel list there is no special column caption for the number of teeth in the "revolution" function.
For the revolution measurement the column "TMO/Overflow" is used to set up the number of teeth. You can find this parameter in the channel configuration dialog under "Advanced Counter Options".
To copy a configuration from one module to another module, the following steps must be carried out:
The measurement software must support XCP-DAQ Packed Mode (XCP 1.4 or later).
If the bitrate of the OUTMM is adjusted CSMconfig cannot find the module despite activated bitrate detection in the settings. The OUTMM is completly passive.
If bitrate detection is activated CSMconfig only listens on which bitrate meanigful information is delivered. Since the OUTMM does not send anything CSMconfig cannot recognize the bitrate and does not find the module.
The OUTMM has limitations at ambient temperature above 85°C.
You can find these additional deviations which may occur in these temperature ranges in the data sheet.
CSMconfig uses a simplified mode for configuration operation. CSMconfig and the XCP-Gateway establish a communication via a special IP adress and a broadcast command.
The measurement software in which the CSM ECAT are integrated does not support this simplified mode. In the measurement software the interface properties (IP adress) must match. If there is a mismatch the measurement sofware cannot communicate with the modules.