!! Please observe the system requirements and the !!
CSMconfig is the configuration software for all measurement devices from CSM. CSM's measurement devices are designed to acquire voltage and temperature signals. The measured values are transmitted via CAN (free running) or Ethernet (EtherCAT®).
PLEASE READ AND UNDERSTAND THE USER MANUAL SHIPPED WITH THE DEVICES BEFORE USING THE DEVICES AND ESPECIALLY BEFORE CONNECTING TO A MEASUREMENT BUS.
The current version of the configuration software CSMconfig supports the following devices:
CSM CAN measurement modules:
ADMM, CNTMM, OUTMM, PTMM, STGMM, THMM
CSM CAN high-voltage measurement modules:
HV THMM, HV TH-TBM, HV ADMM, HV AD-TBM, HV PTMM, HV PT-TBM
CSM CAN ComboModul
CSM EtherCAT® Modules:
AD4 ECAT MM Series, ECAT ADMM, ECAT ADMM HS, ECAT STGMM
CSM EtherCAT® high-voltage measurement modules:
HV AD4 ECAT MM Series, HV IEPE3 ECAT TBM Series, HV STG4 ECAT MM Series
CSM CAN & EtherCAT® high-voltage Breakout Modules:
HV BM 1.1, HV BM 1.2, HV BM 3.1
CSM CAN/SMB xx-Scan measurement modules:
AD-Scan, Thermo-Scan, Dual-Scan
CSM protocol converter:
CSMpressure pressure sensors
ECM exhaust measurement modules:
Klaric measurement modules:
KlariONE, KlariONE Plus
CSMconfig supports several CAN interfaces. A corresponding list can be found in the CSMconfig online help.
Although Intrepid ValueCAN3 is supported, it is recommended not to use this interface. To ensure the proper timing of received messages, the parameters of the ValueCAN driver need to be adjusted to system concerned. These adjustments, however, may interfere with the proper operation of other applications. Please contact Intrepid for details.
Drivers and tools for CAN adapters are not included with CSMconfig.
Win7 / Win8.x /
32- and 64-bit platforms supported
The operating system must be up-to-date. All available updates from Microsoft Update should be installed.
Approx. 100 MB of free space (approximately another 100 MB disk space for CSMview)
Parts of the documentation are delivered as PDF files (Adobe). You need a PDF viewer to read these documents.
PDF-Viewers are available for free, e.g.:
Fixing a runtime error
It may happen that an error message is displayed during installation:
(at -1:0): Cannot import …\RelRegx86.dll
This indicates that components of the operating system required for CSMconfig are missing. This means, the update KB2999226 has to be installed ().
Registration of file extensions
installation, CSMconfig can change the computer-specific default links for the
file extensions DBC and A2L. However, the setup cannot access any user-defined
assignments. These remain unchanged and need to be adjusted manually in the Control
(Default Programs => Associate a file type or protocol with a program).
CSM GmbH regularly provides updates and additional product information. Please visit our website at . Program updates, documents and further files are available for download in the ‘Downloads’ area.
Further product-specific information on CSM measurement technology can be found in the ‘Products’ area.
CSM Computer-Systeme-Messtechnik GmbH
70794 Filderstadt, Germany
Phone:+49 711 - 77 96 40
If a problem cannot be resolved by referring to the information provided, please send a detailed description along with details on your system configuration to CSM. The contact information of CSM's technical support can be found on the website in the 'Support' area.
'Microsoft' and 'Windows' are registered trademarks of Microsoft Corporation. All other trademarks mentioned are properties of their respective owners. EtherCAT® is a registered trademark and patented technology, licensed by Beckhoff Automation GmbH, Germany.
· The latest protocol converter XCP-Gateway 4S pro, equipped with four ECAT inputs and GPS functionality, is fully supported.
· New features for CNT4 evo measurement modules:
o The last counter reading of a selected channel can be stored when the module is switched off. The corresponding option can be enabled in the Channel configuration dialog and the Channel list.
o The module now allows the use of sensor characteristic tables, for example for multi-point calibrations. These tables are specified in the Tables tab of the Device configuration dialog.
o The Revolution function supports additional measurement ranges up to 600,000 rpm and the 32-bit data format.
o The Counter continuous function supports a new measurement range that also allows negative signs. The Zero Adjustment function has been additionally implemented.
· TEDS memories can be overwritten via File | Reconfigure all/TEDS without applying their contents to the configuration document beforehand. The memory content is only checked during readout in CSMconfig, but is not used for configuration purposes. To this end, the Read TEDS dialog has been adapted accordingly.
· In measurement schemes, the column Device name is now available. As a result, device names can be assigned manually regardless of the serial number of the module.
· If devices or channels are disabled, they can be faded out in the configuration document via View | Only active channels or the corresponding icon in the toolbar. The icon changes its appearance depending on the status of the filter.
· If the determination of adjustment values is based on highly noisy raw data, an adjustment may be made to an instantaneous value that does not correspond to the actual offset value. It is thus possible to enable a temporary forced filter in the Auto-Scale wizard.
· The Reload database command in the filter dialog allows the sensor and signal database to be updated while CSMconfig is running.
· The Toggle On/Offline icon in the toolbar changes its appearance depending on the status of the bus connection.
· For a better overview, the documentation in the Help menu is now available in subdirectories, separated according to document type.
· The function to only measure current with HV Breakout Modules on the ECAT side is now activated directly in the Channels selection menu of the Device configuration dialog.
· A new wizard simplifies the configuration of multiple CSMpressure sensors in a measurement chain. Details can be found in the online help under the topic "Integrating CSMpressure".
· Under certain circumstances, measurement ranges taken from the sensor database were not correctly converted into a physical range for STG modules. This has been corrected.
· HV STG4 ECAT measurement modules are supported.
· TH evo measurement modules are supported. In combination with specially developed thermocouple plugs with integrated TEDS chips, these modules provide TEDS functionality (Template 36 and channel names).
· The configuration views Tree View, Device List and Channel List are now combined in a higher-level window to form configuration layouts (=> Window | Select view layout). See CSMconfig help for details.
· Certain AD4 pro CAN measurement modules support module-internal functions for zero adjustment and scaling. See CSMconfig help for details.
· All modules of the AD4 ECAT MM series (as of HW rev. B) now support certain module-internal functions for zero adjustment and scaling. See CSMconfig help for details.
· When saving settings to the module, CSMconfig asks whether internal adjustment values should be retained or reset.
· Several other CAN modules support the simultaneous use of the functions Per channel configuration and Info message. Perform a firmware update to do so. A complete module list can be found in the online help on page "Set CAN ID and Send Rate per Channel".
· The XCP-Gateway Configuration dialog has been improved:
o PTP support is now enabled directly in the PTP tab.
o The XCP tab has been removed. The corresponding settings can be made directly in the Program Settings dialog.
· Shunt and device temperatures are also available as EtherCAT measurement values for the HV Breakout Modules (HV BM). An XCP-Gateway pro is required to be able to use this function.
· If communication between an HV BM and the shunt module fails, this will no longer affect the rest of the measurement chain.
· If the TEDS mode is set to Not using TEDS, all TEDS chips connected will be ignored.
· If an Auto-Configuration has been performed, CSMconfig displays information on the measurement modules, channels and any connected TEDS chips in a message window.
· If a user-defined display range is configured in the Channel Configuration Dialog, this is no longer displayed as a deviation from the TEDS content for LEM sensor packages.
· Sensor names from a sensor database are now also stored in ECAT modules (AD as of HW-Rev. B and STG). Perform a firmware update to do so.
· The handling of multiple XCP-Gateway modules in Vector CANape has been improved.
· When using module-internal adjustment modes, the configuration could not be written to a module if it at the same time contained channel comments.
· If channels were switched off in an AD8 measurement module and then the module-internal adjustment values were written to the module, it could occur that certain values in the unused channels were set to zero.
· When channel comments in the configuration document did not match those stored in the module, they were not correctly listed as deviations in the report.
· TEDS functionality is now also available for modules of the AD4 ECAT MM Series.
· As a supplement to the sensor database, a signal database, containing channel name and comments, can now be specified. These standard names can be assigned to the corresponding channels.
· The sensor database is now also available in CNT evo measurement modules. The functionality corresponds to the measurement scheme and includes the measured variables frequency and revolution.
· The entries in the sensor database and the new signal database are now selected in a separate dialog providing a filtering function (e.g. sensor and channel name).
· CSMconfig can now monitor the calibration date of the connected devices. The calibration date monitoring can be specified in the “Program Settings” dialog.
· During bus scan CSMconfig now automatically checks whether there are firmware updates available for the modules. This function has to be enabled in the "Program Settings" dialog first.
· It is now possible to disable channels on ECAT and CAN modules connected to an XCP-Gateway.
· The sequence of U and I channels can be changed in the HV Breakout Modules. As a result, it is now now possible to measure current only. At the same time, bandwidth can be saved if the voltage measurement is not required.
· The measurement modules ECM LambdaCANc (in 16-bit mode) and CSMpressure can now be used at the CAN interfaces of an XCP-Gateway +CAN.
· Sensor and signal database can be accessed from both the Channel List and the Channel Configuration Dialog.
· If the measurement scheme requires a measurement range that is not available in the AD measurement module used, CSMconfig can automatically set the next larger measurement range and output a corresponding message. This adaption of the settings to device limitations has to be enabled in the "Program settings" dialog.
· To speed up the interface search for XCP-Gateways, this process can be limited to specific networks. To this aim, a positive list of NIC IPs (NIC IP whitelist) can be added in the “Program Settings” dialog. If a whitelist has been set, a corresponding message will be displayed in the “Interface” dialog.
· In the DBC file, the measurement data rate has now been assigned its own attributes. A DAQ software can analyze these to monitor the regular reception of the CAN messages (and thus the measurement data) sent by the CSM measurement modules.
· The designation of the ECAT device types has been adapted to the actual module names and series. This eases the selection of the correct device type for offline configuration. The device names have been modified accordingly.
· The icons in the toolbar and menus have been completely redesigned for more intuitive operation.
· Pressing the Enter key opens the Channel Configuration Dialog in the Channel List view and the Device Configuration Dialog in the Device List view. The previous action (edit cell contents) can now be executed by pressing Shift + Enter.
· In the case of ECM NOxCAN modules, the symbol for enabled channels was still displayed for disabled channels. This has now been fixed.
· Bridge balancing of an ECAT STGMM could reset the module to factory settings under certain circumstances. This behavior is now fixed.
· Support of the new high-voltage Breakout Module HV BM 3.1.
· Certain supported ECAT modules allow the number of channels to be specified to reduce bus load and hide unused channels in the DAQ software.
· Specific AD8 pro measurement modules now provide additional module-internal functions for zero adjustment and scaling. These are very comprehensive features. See CSMconfig help for details.
· PT evo measurement modules now also allow channel comments to be stored.
· Measurement modules TH16 pro and CNT evo now support CAN bitrate 2 Mbit/s.
· A firmware update allows the use of the function "Configuration per channel" for the following new measurement modules:
o HV PT (all)
o ADMM 4 pro HS / AD4 pro MC10 (as of revision F)
· Import and export of measurement schemes:
o Support of measurement module PT8 evo.
o Support of measurement module HV IEPE3 FL1000.
o Support of CNT measurement modules has been extended.
o Serial numbers of currently connected measurement modules are read in during measurement scheme import. These numbers can then be mapped to the corresponding modules in the measurement scheme. This function must first be enabled in the “Program Settings” dialog.
o Measurement schemes contain new columns for Info Message and Device Number.
· When specifying the number of measurement channels for HV Breakout Modules, the corresponding measured quantities are now displayed.
· With the HV BM 3.1, the info message can also be activated when using the "per channel configuration" function. A CAN ID needs to be specified for the info message in order to make use of that feature.
· Selected info message signals (e.g. shunt temperature of HV Breakout Modules) can be included in the A2L file if the CAN modules are connected via an XCP-Gateway.
· As before, the number format for the display of the CAN IDs is by default hexadecimal. The “Program Settings” dialog provides a setting option to change the format to decimal if required. Changing the display mode will apply to the entire user interface and the HTML reports generated.
· The driver library for ETAS BOA interfaces works properly again.
· The CSMcan interface can be used properly again in vMeasure CSM.
· The import of a measurement scheme via vMeasure CSM could cause the program to crash. This has been fixed.
The new HV Breakout Modules (HV BM 1.1 and HV BM
1.2) are supported.
See page “Modules: Using HV Breakout Modules” in the online help file for information on the configuration of these modules.
· Support of HV IEPE3 FL1000 measurement module.
· Support of CSMpressure pressure sensors.
· XCP standard V1.4, which has been supported in CSMconfig for some time, can now also be used in CANape 16 (as of SP4). This standard is required to operate multiple ECAT measurement modules using different measurement data rates >10 kHz simultaneously. It is no longer necessary to set the same measurement data rate for all modules.
· In CSMconfig, various universal cables with integrated TEDS chip (K642 and K833) can be used with STGMM measurement modules for the individual configuration of half and full bridges.
· The channel comment feature is now supported by all high-voltage measurement modules.
· Checking for a new program version is available as an entry in the Help menu.
· Support of the function “Per channel configuration” has been extended to further modules. Suitable firmware updates are now available for the modules listed below and included in CSMconfig:
o ADMM 8 pro (Rev. E)
o THMM 16 pro (Rev. D)
o THMM 8 pro (Rev. G)
o THMM 8 pro (Rev. H)
A complete list of all supported modules can be found in the online help file on page “Setting of CAN ID and send rate per channel”.
· Vector ETH interfaces support via Vector DAQ software (e.g. CANape 16 as of SP4) has been extended.
· If the automatic TEDS mode is used, CSMconfig now detects a TEDS which has been connected to a wrong input. If the serial number of the connected TEDS chip does not match the one stored in the configuration document, the symbol of the corresponding measurement channel is displayed in red. This also causes an error message in the Vector DAQ software and prevents measurements with incorrectly plugged sensors. The TEDS mode can be selected in the “Program Settings” dialog.
· Columns in list views are now sorted correctly according to the data content. This means, for example, that columns containing numeric values are now sorted strictly numerically.
· Measurement modules and channels can now be enabled and disabled directly by using the "Active" column in the Device and Channel list. The user can also select multiple modules and channels simultaneously.
· The module-internal zero adjustment can be called up in the Channel Configuration Dialog via the new button "Zero Adjust.". This button is only available if the function is supported by the connected measurement module. In order to make use of this feature, the option "Allow module internal zero adjustment" has to be enabled beforehand in the “Program Settings” dialog.
· CSMconfig now provides the means to insert scaling texts into A2L files. This option must be enabled in the "Program settings" dialog if required.
· Template files are now backward-compatible. Measurement modules that are not yet supported in the CSMconfig version currently used are ignored.
· OUTMM modules can now be properly operated again when using the CAN interfaces of an XCP-Gateway +CAN.
· When loading a DBC file, it could occur that the two-point scaling of CNT MiniModules was not correctly read out. That error has been fixed.
· When using the Motorola format and the option “Per channel configuration" at the same time, the start bit of the CAN message is now correctly specified.
· If the XCP compatibility is set high in the program settings, PTP is no longer automatically activated in the XCP-Gateway.
· Under particular circumstances, deleting modules from a configuration document could cause the program to crash. This has now been fixed.
· CSMconfig creates a full data backup before performing a firmware update. This ensures that all information stored in the module (e.g. for CAN modules also the date of the last calibration) is kept reliably.
· CSMconfig provides options to define module- and program-specific defaults and to save them in template files.
· For configuration documents: Default settings can be defined separately according to document type (*.DBC/A2L) for every module type (e.g. name pattern).
· For program-specific settings: Program settings can be defined according to user-specific requirements and saved in a template file (*.creg).
These default settings can be used as templates for installation on multiple computers (e.g. for a company-wide installation of CSMconfig) and can be transferred to all computers simultaneously.
· As of firmware version 8.3, the output of specific measured quantities in 16-bit format is supported by ECM LambdaCANc. A mixed operation mode, i.e. the output of 16- and 32-bit values is possible.
· At each program start, CSMconfig checks whether a new program version is available. This procedure can be configured and switched off.
· Saving documents in the old DBC file format (i.e. older than CSMconfig V8.5.0) is no longer supported.
· The option Write Name to TEDS has been added to the Channel Configuration Dialog. To use this option, it first needs to be enabled in the Program Settings dialog (Allow adding channel name to TEDS).
· CSMview V2.0.0 is an integral part of the installation package and is by default installed along with CSMconfig. This program version also supports ECAT and CAN measurement modules connected via an XCP-Gateway.
· Vector Ethernet interface VX0312 is supported.
· The icons for modules and channels have been redesigned for better usability. Especially the status indication for TEDS sensors has been improved.
· The Temperature Coefficients tab in the Channel Configuration Dialog for PTMM modules has been revised and made clearer.
· The XCP-Gateway Configuration dialog has been enhanced. It now also displays the NIC mask of the network interface used. By pressing the button Adjust IP to my NIC, the valid IP settings of the NIC configuration can be read out.
· CAN measurement modules can be operated at a maximum measurement data rate of 1 kHz via an XCP-Gateway +CAN. Higher measurement data rates are no longer provided for configuration. Until now, only an error message was displayed, when saving the configuration.
· As for STGMM modules, the setting of the filter is now also taken from the TEDS.
· When trying to open another instance of CSMconfig, the current window will appear in the foreground.
· In order to speed up the program installation, the setup now automatically detects software packages already installed on the system.
· A channel list is no longer created all over again when activating previously disabled measurement channels. Only the lines required will be re-written.
· Stored column layouts are migrated to new program versions complete and unchanged.
· The device list is now also updated when duplicating modules (copy and paste) in Tree View mode.
· The Tree View of a configuration document will be updated if multiple lines of a channel list have been selected and modified.
· For some CAN measurement modules, CAN ID and measurement data rate can be specified per channel (option Per channel configuration). This also works correctly now for CAN measurement modules connected via an XCP-Gateway +CAN.
· In the Device Configuration Dialog, a change in the number of channels used to overwrite a changed measurement data rate if the latter had not been previously saved. This error has now been fixed.
· CSMconfig verifies the calibration date stored in the TEDS. The program settings provide an option to define, how many days before the calibration expires a warning alert will be displayed. Negative values are no longer permitted.
· The text input fields only allow characters matching the language settings of the computer.
· CSMconfig will reliably detect if changes have been made in the XCP-Gateway Configuration dialog and written to the device. The configuration document can be saved again.
· Some dialogs will close automatically after a pre-defined period of time. As long as such a dialog is visible in the foreground, no changes can be made in other program components currently located in the background.
· Buttons in tables (e.g. channel lists) are no longer obscured by tooltips.
· The XCP-compatibility for Vector CANape 15 now corresponds to XCP 1.3.
· If the busload is exceeded during the configuration of an ECAT measurement chain or an individual ECAT module, a warning message will be displayed. This helps to detect and avoid overload configurations.
· The busload of an ECAT measurement chain is now displayed in the report of a configuration document. The figure is displayed below the XCP-Gateway entry in percent relative to the permitted maximum.
· Measurement schemes using the CAN identifier “0” can now be imported.
· A bus load issue concerning the CAN ports of the XCP-Gateway has been fixed.
· An uncritical syntax error which occurred in newly created A2L files when a sensor database was used has been fixed.
· The property Device Number is not supported by ECAT STGMM modules. The corresponding field in the Device Configuration Dialog therefore appears greyed out now.
· When using TEDS sensors on ADMM modules, a deviating configuration is highlighted in color now. This now also works correctly for the property Range in the Channel Configuration Dialog.
· When using TEDS sensors on ADMM modules, it could occur in exceptional cases that specific measurement ranges were not correctly read out from a TEDS and transferred to the channel settings. The error has been fixed.
· The standard filter in ECAT modules is now correctly displayed when the measurement data rates 500 kHz and 1 MHz are used.
· When using a sensor database, specific window sizes of the views (especially Channel list) could cause the program to seem to freeze when opening individual dialogs. The error has been fixed.
· HV AD4 ECAT MM series supports all specified measurement data rates.
· Bridge balancing of STGMM works correctly even when using sensor cables with integrated TEDS chip.
· Measuring unit from the TEDS chip of the quarter bridge extension cable (K356) for STGMM is being read correctly.
· With ECAT ADMM HS (hardware revision Axxx) scaling values (factor/offset) are correctly written to and read out from the device (internal calibration data are handled correctly).
· If several parameters in the device dialog of a module are changed simultaneously, malfunctions may occur. In this case the module should be restarted and the settings should be made or written one after the other.
· Filter settings read from a TEDS connected to STGMM (1/4 CSM) will be applied only if the parameters are compatible with the rate setting of the module. Otherwise, the default filter is used.
· The detection of changes on connected TEDS sensors works delayed on most modules. This can result in a TEDS change not being properly updated in the GUI. In this case repeat the bus scan at a time interval (>10s) or restart the module.
· Some new program options from V8.7.0 were not correctly saved.
· TEDS functionality for CAN-based ADMM modules. With TEDS-capable ADMM modules, channel configuration data can now be transferred from the TEDS to the channel configuration.
· XCPG/PTP: It is now possible to set the end-to-end or peer-to-peer delay mechanism. To make use of this function, firmware version 3.10 or later needs to be installed on the XCP-Gateway.
· Integration of a sensor database for simplified configuration
· Version check of column definitions for the import/export of measurement schemes
· Support of ASAM MCD-1 XCP V1.4 according to specification as of 2017-05-24
· Integration of new modules
o AD 8 pro MC2 as of hardware revision H4xx: Further development of ADMM 8 pro featuring TEDS functionality
o AD 8 pro2 MD2 as of hardware revision G4xx: Further development of ADMM 8 pro2 featuring TEDS functionality
o HV AD4 XW1000: New development of an ECAT-based HV AD MiniModule with a measurement data rate of max. 1 MHz and a measurement range up to 1000 V
o HV AD4 OW1000: New development of an ECAT-based HV AD MiniModule with a measurement data rate of max. 1 MHz and a measurement range up to 90 V
o HV AD4 XW20: New development of a CAN-based HV AD test bench module with a measurement data rate of max. 20 kHz and a measurement range up to 1000 V
· New setup features
o Uninstallation of previous program versions
o Registration of file extensions DBC and A2L
· Improved compatibility to some specific firewalls
· xx-Scan modules no longer need to be restarted after configuration in order to send measurement values.
· OUTMM test mode: Default output values now refer to the electric output variable.
· The inclusion of stuff bits allows more precise bus load calculations.
· When calculating the bus load, every module is individually checked in addition to the verification of the entire configuration in order to avoid overload configurations.
· CNTMM: When activating the virtual channels, all parameters are now properly adopted from the real channels.
· A document is no longer opened twice when clicking on the corresponding file shortcut.
· The Open file dialog only lists files of the type defined in the Program Settings dialog (DBC/A2L).
· The auto-configuration of an XCP-Gateway CAN interface in the Databases tab of the XCP-Gateway Configuration dialog is functional again.
· The default rate of CAN-based STGMM modules is now 100 Hz instead of 1 kHz. This prevents overload configurations when a device is reset to factory settings.
· Improved validation of IP configuration when saving A2L files
· Integration of new modules
o ECAT ADMM 4 HS10/100/1000 (HW-Rev. Bxxx)
o HV AD4 XW20 (CAN-based HV measurement module for operating voltages up to 1000 V)
o ADMM 4 (HW-Rev. Bxxx) a new version of our smallest AD measurement module with CXS housing
· THMM 8/16 classic measurement modules now support customer-specific measurement data rates.
· Synchronization of multiple XCP-Gateways via IEEE1588 V.2 (PTP)
o Firmware version 3.06 or later is required in order to enable the PTP functionality in an XCP-Gateway
o Extended menu functions for the configuration of PTP parameters
o Adaptation of A2L files as from XCP version 1.3
· Simultaneous change of multiple parameters in device and channel lists possible
· Resizable dialog windows
· Error messages displayed when saving an A2L file are now available in German.
· Simplified copy-and-paste / marking operations in device and channel lists
· If an automatic configuration is performed, CSMconfig checks the compatibility of the interface with the document (CAN/EtherCAT®).
· If a measurement scheme import is performed and UserMin > UserMax, these values will be exchanged against each other.
· An error that occurred when saving a DBC file has been fixed.
· Improved compatibility of generated A2L-files to DAQ-software.
· Extensions in configuration via measurement schemes:
o Interpolation tables of ADMM ‘pro’ modules are now also supported.
o Support of measurement scheme import and export for the Combo Module.
· Support of a maximum measurement rate of 20 kHz for the upcoming module ECAT STGMM 6 as of hardware revision B013 and firmware V2.02. Operation is possible in combination with the following XCP-Gateways:
o XCP-Gateway with hardware revision B and firmware as of V2.30
o XCP-Gateway with hardware revision C and firmware as of V3.06
· Prepared for new measurement mode: signals can be configured to be transferred on separated CAN-IDs per channel for the following modules:
o THMM 8 pro with hardware revision G,
o THMM 16 pro with hardware revision D,
o ADMM 4 pro or ADMM 8 pro resp. with hardware revision F,
o ADMM 8 pro2 with hardware revision E.
To use this feature, the modules require a firmware update that will be available soon.
· Updated firmware versions for several modules available:
o STGMM 6, HW-Revision A: V1.32
o HV TH-TBM 8, HW-Revision A: V1.07
o OUTMM, HW-Revision A: V1.34
o ECAT STGMM 6, HW-Revision A and B resp.: V1.32 and V2.02 resp.
o XCP-Gateway, HW-Revision B and C resp.: V2.30 and V3.06 resp.
· Changing the baud rate is no longer allowed in setups containing deactivated modules in order to prevent a potential overload of the CAN bus.
· For ADMM 8 and ADMM 8 pro modules (also ComboModul) with hardware revision D and F, a new firmware version V4.54 and V6.09 resp. is available. We strongly recommend to update these modules.
· Accelerate: the time needed to configure and adjust ECAT measurement modules in large-scale setups is significantly reduced.
· Clear view: the size of several dialogues e.g. for firmware update or measurement scheme import can be adjusted.
· The device list now also shows the hardware revision of the modules.
· Configuration of OUTMM modules that are controlled by modules connected to the same CAN port is possible now also via XCP-Gateway.
· Improved consistency: a new dialog prevents ambiguities in device configuration when closing CSMconfig or a view.
· Handling when working with interpolation tables is improved
· Updated firmware versions for several modules available
· The default filter settings and the unit for send rate to be used for measurement scheme import can be selected in the import dialog.
· Errors which occurred when using the function “Duty Cycle” in the CNTMM channel configuration dialog have been fixed.
· Updated firmware versions for several modules available
· The DBC file describing the control signals for an OUTMM was accidentally written in Unicode format.
· Adjustable PT coefficients for the upcoming modules PTMM 4 evo and 8 evo as well as for HV PTMM 2 and HV PT-TBM 8.
· Support of upcoming modules OUTMM with hardware revision B and THMC 8 with hardware revision B.
· Support of the ComboModule
· Support of filter type Median 11 on THMM8 pro
o Modules with HW revision F and firmware as of V6.04
o Modules with HW revision G and firmware as of V7.16
· The column layout in channel and device list view can now be easily modified
· Simplified columns selection in the channel and device list view, the layouts can be exported and imported.
· Rearrangement of channel configuration dialogs
o As already implemented in the STGMM channel dialog, tabs for various parameter groups are now also available for other CSM modules
o The current measurement value is displayed in the channel dialog
· Revised dialog for device type selection (structured according to device classes)
· New dbc format: channel comments can now be read by MC software (e.g. INCA). As of version 8.5.0, CSMconfig still reads the former dbc format but uses per default the new format for writing the configuration.
· General performance improvements, especially for large-scale setups and in combination with TEDS sensor cables or adaptors.
· Improvements in OUTMM test mode to avoid the sporadic output of default values.
· ECM modules NOxCANg and NOxCANt will now be detected when a bus scan is performed.
· Operating problems of the function ‘Reconfigure all’ and reading of module name in combination with ECM modules have been fixed.
· Incorrect dependencies of channel parameters in ‘Autoscale Wizard’ fixed.
· Resetting the modules to factory values now also includes channels that have not been used.
· Stability improvements during firmware update for ECAT measurement modules.
· Incomplete storing of channel configuration in TEDS of a connected quarter bridge completion cable fixed.
· Status messages indicating a timeout condition modified for CNTMM 4 modules.
· As from version 8.4.0, CSM’s configuration software, formerly known as xx-Scan Config, is named CSMconfig. The version numbering continues and the tool’s graphical user interface remains the same. Also, CSMconfig continues to feature all the functions and benefits the user is familiar with.
· The online help is available in German.
· A new operating mode in XCP-Gateway allows to operate up to 100 ECAT measurement modules at one XCP-Gateway (as from firmware V2.26). Up until now, only 25 ECAT measurement modules could be connected to one gateway.
· Performance optimization by buffering: A module’s configuration information is read out once and written in the RAM of the computer. This speeds up the access time during the configuration process.
· Simplified configuration of STGMM measurement modules: The configuration of a measurement channel can be directly written to the device from the channel configuration dialog. The effect on the measured value is displayed in the channel configuration dialog.
· Improved consistency for STGMM channels: When using quarter bridge completion cables, the entire configuration of the measurement channel can be stored in the completion cable’s TEDS. For ECAT modules, firmware V1.28 is recommended
· Simplified user navigation for the configuration of XCP-Gateway and ECAT MiniModules
· Improved measurement scheme export for STGMM measurement modules (CAN and ECAT)
· As from version 8.4.0, measurement modules can no longer be connected via serial interface.
· CSMconfig now correctly compares filter settings for all module types with the filter settings stored in a measurement module.
· When measurement modules are reset to factory settings, previously deactivated channels will reactivated.
· The device properties of ECAT measurement modules are read out correctly by the XCP-Gateway.
· Send rate settings that could lead to a bus overload are prevented.
· V8.3.1 is a special version of CSMconfig for a customized version of ECAT ADMM HS 800. Details on configuration and operation are available as Technical Note.
· German GUI version for CSMconfig (online help still only in English)
· Extended and updated documentation
· Refresh rate for the visualization of measurement values has been enhanced
· Improved user-friendliness: The progress of longer lasting processes (e.g. auto configuration of comprehensive measurement chains) is indicated in a small status window.
· Simplified configuration of measurement channels: Measurement channels can be individually reset to factory settings via the <Defaults> button in the channel configuration dialog.
· Simplified handling of configuration files: When saving a file, the previously selected directory path will be used (this also applies for CSV files used for the import of measurement schemes).
Enhanced visualization of measurement data
during CAN measurements:
The current values of slowly sending measurement modules are fully displayed.
· Import of foreign DBC files
o Enhanced import process.
o The import of a foreign DBC file into a regular CSM DBC file can be completely reversed via the command “Remove foreign devices”. The contents of both files will be completely separated from each other.
· When configuring ADMM pro modules, it is now possible to delete individual scaling tables via the “Delete” command in the context menu.
· Support of the soon available high-voltage measurement modules HV ADMM 4LI, HV ADMM 2+, HV AD-TBM 4+ /65, HV AD-TBM 8LI /65, HV PTMM 2 and HV PT-TBM 8 /65
· Improved configuration of XCP-Gateway +CAN:
o When opening an A2L file, the user will be prompted to open the contents of the CAN DBC file included in the A2L file in a separate window.
o If a CAN DBC file included in an A2L file is modified, the user will be prompted to save the changes before closing the A2L file.
o Clearly arranged display of XCP-Gateway license features.
· Simplified measurement procedures for CAN STGMM 6 pro HS and ECAT STGMM 6:
o The polarity of quarter and half bridge strain gauges can be reversed via software (as from firmware v1.25).
o If a filter is used for measurement, its settling time will be taken into account when bridge balancing is performed.
o Bridge balancing can be triggered by a CAN message (only CAN STGMM 6 pro HS and firmware v1.25 or higher)
· The “Auto-Configuration” shortcut has been changed from CTRL + A to ALT + A to solve a conflict with the Windows shortcut CTRL + A (Mark all).
· When saving an A2L file, CSMconfig checks the network settings (IP configuration) of the XCP-Gateway.
· Simplified configuration of CAN-based measurement modules, connected to the XCP-Gateway.
· Supports the module ECAT ADMM 4 HS100 with galvanically isolated sensor excitation.
· The latest firmware update (version V4.35) for AD-Scan and Thermo-Scan modules is integrated.
· The latest firmware update (version V1.23) for the module ECAT STGMM 6 Rev.A000 is integrated.
· The latest firmware update (version V1.06) for the module ECAT ADMM 4 HS Rev.A000 is integrated.
· The firmware (version V4.01) for the module THMM 16 Rev.D000 is integrated.
· The firmware (version V7.12) for the module THMM 8 Rev.G000 is integrated.
· Like the ECAT version STGMM 6, the CAN module STGMM 6 pro HS now also supports the storing and read-out of bridge balancing results to/from the cable if a strain gauge connection cable with built-in quarter bridge completion is used (K356).
· For the sake of clarity, the scaling formula in the channel dialog now includes the unit of measurement.
· MiniModules supporting the baud rate 2 Mbit/s are now detected by the feature “bitrate detection”.
· For users with limited access, CSMconfig can be installed in a read-only mode. This limited access does not allow the configuration of CSM measurement modules and read-in DBC files cannot be modified.
· Support of the XCP-Gateway CAN interfaces in CSMconfig (requires XCP firmware v. 2.20 and licensing of CAN interface). This allows CAN bus bitrates up to 1 Mbit/s and sampling rates between 10 Hz and 1 kHz.
· ECAT measurement modules STGMM and ADMM: The filter setting “Std/300” Hz has been erroneously read out
· Integration of CSMview for verifying measurement configurations.
· If a cable for quarter bridge completion (K356) is used in combination with an ECAT STGMM 6 measurement module, the bridge balancing result can now be written to the cable and read in from the cable.
Support of the new high-speed measurement module
ECAT ADMM 4 HS800.
For this purpose, the firmware for the XCP-Gateway needs to be updated at least to version 2.20. Also, the validity of the license needs to be checked and – if required – updated.
XCP-Gateway now also supports CAN-based
For this purpose, the firmware for the XCP-Gateway needs to be updated at least to version 2.20. Also, the validity of the license needs to be checked and – if required – updated.
o The CAN functionality of the XCP-Gateway +CAN is limited at present. Please, see the online help for more information.
CSM reserves the right for changes without
© CSM GmbH, Filderstadt, Germany